Arch Koh Adapter Cleaning and Sterilizing Instructions
Remove the inner rod from the outer arm.
Disconnect the Arch Koh Adapter from the inner rod assembly
at the ball joint and remove the joining screw from the
Arch Koh Adapter. (See pg. 7 of the Arch Koh Instruction
Inspect parts for defects, such as parts not fitting together
or misshapen parts. Contact Side-Kick sales reps. regarding
Use Enzol® enzymatic detergent, and prepare according to
the manufacture's recommendations of 1 oz. per gallon of warm
tap water, or an FDA approved equivalent.
Soak the outer arm, inner rod and Arch Koh Adapter for five
minutes in the prepared solution.
Maneuver movable parts to loosen trapped soil. Use a soft
bristled brush (M16) to gently scrub to remove any visible
soil and actuate all moveable parts while scrubbing in order
to loosen trapped soil. Use an appropriate size lumen brush
(45-541) for internal channels.
Repeat steps 5 and 6 two addional times for a total of three
Prepare a tap water bath at 38-49℃.
Agitate the assembly by hand for a minimum of one minute in
the prepared bath.
Rinse the assembly under running tap water for at least one
Dry the exterior of the assembly with a clean, lint free cloth.
Place items in the Kronner Sterilizer tray.
Sterilization wrap: Kimgard KC400 or FDA approved equivalent
Pre-vacuum temperature: 270℉ (132℃)
Exposure time: 4 minutes
Drying time: 45 minutes